to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||23 March 2007|
|PDF File Size:||10.59 Mb|
|ePub File Size:||4.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
Caution, consult accompanying documents.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Page 55 Performing a measurement The three-digit code number which is also printed on the reverse side of the code strip and on the mqnual strip container is shown in the display.
If the problem persists, please contact your local customer support and service center. Press the M button to change the currently flashing figure and to set the current year.
Quality control recommendation Applying the control solution 10 Apply a large free hanging drop of either the glucose or cholesterol low level control solution directly from the bottle to the test strip. Medical staff and other people using the Accutrend Plus instrument to perform measurements for more than one patient must be aware that any object coming into contact with human blood is a potential source of infection.
Inserting The Code Strip Performing a measurement Inserting the code strip Hold the code manaul with thumb and index finger in the white area located at the end of the strip. If you are measuring glucose, perform the plausibility check described on the next page.
Cobas Accutrend Plus User Manual
If the measured val- ues are to be stored with time information, enter the correct settings see the Setting the date section of this manual. Quality control recommendation Starting the measurement 11 Close the measurement chamber flap. Quality control recommendation The measurement time varies depending on the pls parameter. Performing a measurement Hold the test strip with thumb and index finger so that the sample application area is facing upwards.
Quality control recommendation After switching on the instrument, check the following displays: To run a test, an unused test strip is taken from the container and inserted into the instrument.
Quality control pllus Now take the test strip from the test strip container. Make a habit of carrying out a function check in the following situations: Page 3 The packaging material, the identification plate of the instrument and the manual may contain the following symbols or abbreviations which are listed below with their meaning: Page 49 Performing a measurement Test strips Code strips Quality Control Recommendation Quality control recommendation Quality control recommendation To ensure that the instrument is functioning properly, you should regularly use two levels of control solutions for a function check.
Also always check the battery condition after the display test. Accurend a set of fresh batteries, you will normally be able to perform approximately 1, measurements.
Performing a measurement Switching on the instrument Place the instrument on a level surface or hold it in your hand. Quality control recommendation Display of results When measurement is finished, the result is displayed. Memory To navigate within p,us memory area, use the M button. Contents Product specifications Operating conditions and technical data If any problems occur during coding, the meter displays an error message see Error Messages chapter of this manual.
The display then automatically switches to the mode for setting the current day. Error messages Error messages In certain circumstances error messages may appear on your display.
Page 96 Memory Special cases: We recommend that you always leave the beeper enabled. Cleaning Cleaning Cleaning A clean optical measuring system is a basic prerequisite for determining precise measured values.
Performing a measurement Blood application in the instrument: Performing a measurement Sample material Fresh capillary blood accuteend used as sample material. Flap open displayed after instrument has been powered on The measurement chamber flap is not closed properly. Insert the code strip smoothly into the test strip guide up to the stop in the direction of the printed arrows. Contents Quality control recommendation Preparing to perform a control test Measurement results which fall outside the meter measurement range are displayed as Hi above the measurement range or Lo below the measurement range.
Manufactured for and distributed in the U.
Are date and time correct? Starting The Measurement Performing a measurement Starting the measurement 13 Close the measurement chamber flap. Operating the instrument Ensure the instrument is switched off and turn it over. Page 69 Performing a measurement Blood application in the instrument: Repeat the procedure with a new strip.
Preparing To Perform A Control Test Quality control recommendation Preparing to perform a control test Get the test strip container for the required measurement e.